19 October 2009
Intraocular pressure measurements with the newly reconfigured Ocuton S*TT-MV self-tonometer in comparison to Goldmann applanation tonometry in glaucoma patientsElena LanfermannBCDEF, Clemens JürgensCDE, Rico GroßjohannCDE, Susanna AntalAB, Frank TostACDEFG
Med Sci Monit 2009; 15(11): CR556-562 :: ID: 878239
This study compared the new Ocuton S*TT-MV to the Ocuton S (former configuration) and GAT. The effect of the reconfiguration on measuring accuracy and device handling of glaucoma patients was evaluated.
Material and Method
101 glaucoma patients performed self-measurements using the old Ocuton S and the Ocuton S*TT-MV. The new Ocuton S*TT-MV automatically verifies the applanated area. We used GAT as the gold standard reference. Three series of measurements were performed in random order.
The mean IOP was 18.3+/-4.2 mmHg with Ocuton S, 17.5+/-3.6 mmHg with the reconfigured Ocuton S*TT-MV, and 15.1+/-3.4 mmHg with GAT. IOP ranged from 9 to 30.5 mmHg with GAT, 7 to 42 mmHg with Ocuton S*TT-MV, and 5 to 38 mmHg with the old Ocuton S. The mean difference between the old Ocuton S and GAT was 3.2+/-3.5 mmHg, between Ocuton S*TT-MV and GAT 2.4+/-3.3 mmHg, and between old and new Ocuton S 0.9+/-2.9 mmHg (p<0.01). The standard deviation within an individual row of 4 readings per eye with the old Ocuton S averaged 1.9 mmHg, while with the new Ocuton S*TT-MV the standard deviation averaged 1.7 mmHg. This measuring accuracy showed no age correlation for either self-tonometer (old Ocuton S: coefficient of correlation R=0.2; Ocuton S*TT-MV: coefficient of correlation R=0.053).
The newly reconfigured Ocuton S*TT-MV improves measuring accuracy significantly and approaches the accuracy achieved using GAT. Improved usability of self-tonometers has clinical relevance, as patients don't have to check the corneal cast on the measuring prism. This could lead to an increased usage of self-tonometry in glaucoma monitoring.
Keywords: Intraocular Pressure - physiology, Glaucoma - physiopathology, Cornea - pathology, Astigmatism, Age Distribution, Self Care, Tonometry, Ocular - instrumentation
18 October 2021 : EditorialEditorial: Global Initiatives Support the Use and Regulation of Digital Health Technology During the COVID-19 Pandemic
Med Sci Monit 2021; 27:e935123
11 October 2021 : Clinical ResearchClinical Features and Temporal Lung Radiographic Changes in 25 Patients Recovering from COVID-19 Pneumonia:...
Med Sci Monit In Press; DOI: 10.12659/MSM.933381
08 September 2021 : Clinical ResearchAcceptance of COVID-19 Vaccination and Its Associated Factors Among Cancer Patients Attending the Oncology ...
Med Sci Monit In Press; DOI: 10.12659/MSM.932788
18 October 2021 : EditorialEditorial: Global Initiatives Support the Use and Regulation of Digital Health Technology During the COVID-...
Med Sci Monit 2021; 27:e935123
18 Oct 2021 : Clinical ResearchAdverse Side Effects Associated with Corticosteroid Therapy: A Study in 39 Patients with Generalized Myasth...
Med Sci Monit In Press; DOI: 10.12659/MSM.933296
13 Oct 2021 : Clinical ResearchEndovascular Repair of Thoracic Aorta Injury: 17 Years of Single-Center Experience
Med Sci Monit In Press; DOI: 10.12659/MSM.934479
11 Oct 2021 : Clinical ResearchChallenges for Polish Community Pharmacists in Provision of Services to Immigrants and Non-Polish-Speakers ...
Med Sci Monit In Press; DOI: 10.12659/MSM.933678
Most Viewed Current Articles
20 Mar 2020 : Clinical ResearchSocial Capital and Sleep Quality in Individuals Who Self-Isolated for 14 Days During the Coronavirus Diseas...
Med Sci Monit 2020; 26:e923921
15 Apr 2020 : Clinical ResearchPsychological Impact and Coping Strategies of Frontline Medical Staff in Hunan Between January and March 20...
Med Sci Monit 2020; 26:e924171
05 May 2020 : Review articleAn Evidence Based Perspective on mRNA-SARS-CoV-2 Vaccine Development
Med Sci Monit 2020; 26:e924700